|ACS Instrument Handbook for Cycle 26|
4.4 4.4.1 MAMA PropertiesThe ACS MAMA detector is the STIS flight spare STF7. It provides coverage from 1150 Å to 1700 Å. The MAMA detector is a photon-counting device which processes events serially. The ACS MAMA only operates in the accumulate (ACCUM) mode, in which a time-integrated image is produced. Unlike the STIS MAMAs, ACS does not offer the high-resolution (2048 × 2048) mode or time-tagged data acquisition. Figure 4.13 illustrates the design of the MAMA, which has an opaque CsI photocathode deposited directly on the face of the curved microchannel plate (MCP). Target photons strike the photocathode, liberating single photoelectrons which pass into the MCP, where a pulse of ~4 × 105 e– is generated. The pulse is recorded by an anode array behind the photocathode and detected by the MAMA electronics which rejects false pulses and determines the position of the photon event on the detector.The field electrode, or repeller wire, repels electrons emitted from the microchannel plate back into the channels. This provides an increase in quantum efficiency of the detector at the price of an increase in the detector point spread function halo. The repeller wire voltage is always on for SBC observations.The ACS MAMA has a broken anode which disables rows 600 to 605. There are three dark spots smaller than 50 microns at positions (334,977), (578,964), and (960,851), as well as two bright spots at (55,281) and (645,102) with fluctuating rates that are always less than 3 counts per second. The reference pixel has been moved to (512,400) to avoid these areas (see Table 7.10)Figure 4.13: Design of the SBC MAMA.Table 4.8: SBC detector performance characteristics.
1024 × 1024 pixel 25 × 25 μm2 ~0.034 × 0.030 arcsec2/pixel 34.6 × 30.5 arcsec2 Dark count1 8.11 × 10–6 counts/s/pixel Global count-rate linearity limit2 < 1.2 × 10-9 above 400 nmThe dark current remains constant up to an instrument temperature ~25°C. Above that temperature the dark current increases rapidly.
4.4.2 SBC Spectral ResponseThe total transmission curve for the SBC with the PR110L prism is shown in Figure 4.14. The peak photocathode response occurs at Lyman-α. Its spectral response is defined by the cutoff of the MgF2 window at 1150 Å at short wavelengths, and by the relatively steep decline of the CsI photocathode at long wavelengths.Observations of flux calibration stars and a G-type star using the SBC PR110L prism have revealed that the sensitivity of the MAMA detector to optical and near-UV light is apparently much larger than previously thought ( ACS TIR 2008-02).Estimates of the real SBC throughput indicates that the detector efficiency is factors of approximately 50 and 1000 higher at wavelengths of 3000 Å and 4000 Å respectively compared to ground testing. For a solar type spectrum, this can mean that one-half or more of the counts detected are due to optical and near-UV photons, rather than from the expected FUV photons. The updated sensitivity curve has been incorporated both in the Exposure Time Calculator (ETC) and in pysynphot. There is also some evidence that this red leak changes as the SBC detector warms up, increasing by as much as 30% over the course of 5 orbits. It is not yet clear if this red leak has also been increasing secularly over time. STIS FUV MAMA data seem to show a similar, although perhaps somewhat smaller effect.For dispersed PR110L and PR130L observations, it is straightforward to identify this extra red light; however, it clearly also affects SBC imaging observations done with the long pass filters. Until this effect is better understood and calibrated, extreme caution should be used when interpreting FUV imaging observations of red targets. Observers who need to measure FUV fluxes of red targets are strongly recommended to take interleaving observations with two different SBC long pass filters (e.g., F140LP and F165LP), so that the difference in the count rates can be used to isolate the true FUV flux. There is some recent evidence that the red leak is temperature dependent and cannot be accurately determined in the F165LP filter. For this reason it is recommended that this filter not be used for science. It should only be used in conjunction with other filters to subtract red photons from the measurements in other bandpasses.4.4.3 Optical PerformanceThe SBC exhibits low-level extended wings in the detector point-spread function (PSF) (ACS ISR 2016-05). Sample MAMA detector PSF profiles are shown in Figure 4.15.