In this mode the WFC CCD accumulates signal during the exposure in response to photons. The charge is read out at the end of the exposure and translated by the A-to-D converter into a 16 bit data number (DN), ranging from 0 to 65,535. The number of electrons per DN can be specified by the user as the GAIN value. The full well of the WFC CCD is about 85,000 electrons and consequently, all GAIN values larger than 1 will allow the observer to count up to the full well capacity. For GAIN=1
only 75% of full well capacity is reached when the DN value saturates at 65,535. The readout noise of the WFC CCD is about 4 electrons rms and thus it is critically sampled even at GAIN=2
. WFC can make use of a user-transparent, lossless, on-board compression algorithm, the benefits of which will be discussed in the context of parallel observations. The algorithm is more effective with higher GAIN values (i.e., when the noise is undersampled). Note that only GAIN=2 is supported by STScI.
Several supported apertures (see Table 7.6
) are accessible to WFC users. WFC1-FIX
select the geometric centers of the two WFC camera chips. WFCENTER
corresponds to the geometric center of the combined WFC field, and will be useful for facilitating mosaics and obtaining observations at multiple orientations. Due to maximal CTE loss, WFCENTER is not recommended for a single compact target. WFC, WFC1,
are located near the field of view center and the centers of chips 1 and 2, respectively (see Figure 7.3
), Their locations were chosen to be free of detector blemishes and hot pixels, and they are the preferred apertures for typical observations. See Section 7.7
for more details about ACS apertures, including subarray apertures.
The present flight software does not allow reading an ACS frame directly into the HST
on-board recorder. Images must first be stored in the internal buffer. When more than one WFC full-frame image is obtained during an orbit, a buffer dump must occur during the visibility period so as to create space in the buffer for a new WFC image.
To select the desired wavelength, the ramp filter is rotated to move the appropriate part of the filter over the specified pointing. Observations with different ramp filters do not generally occur at the same pointing. The precise location where a given observation will be performed can be found from Table 7.6
where for each ramp filter we list the fiducial pointing for the inner IRAMP, middle MRAMP, and outer ORAMP filter segment. The inner segment corresponds to the WFC1 CCD, while the outer segment corresponds to the WFC2 CCD. The middle segment can be used with either of the WFC CCDs but only the WFC1 aperture is supported.
For any ramp filter observation three ramp filters will end up in the FOV even though the target is properly positioned only for the requested one. Table 5.1
and Table 5.2
can be used to determine if the remaining two ramp filter segments are useful for serendipitous observations. The user can request either full-frame readout, or the 2K subarray readout containing only the primary ramp filter's illumination region.
The SBC ACCUM
mode accumulates photons into a 1024 by 1024 array, with 16 bits per pixel. The data are sent to the onboard recorder via the internal ACS memory buffer at the end of the exposure. ACCUM
is the only mode available for SBC observations; the Time Tag mode of the STIS MAMAs is not available on ACS. The minimum SBC exposure time is 0.1 second and the maximum is 1.0 hour. The minimum time between SBC exposures is 40 seconds. Note that the SBC, like the STIS MAMAs, has no readout noise. As a consequence there is no scientific driver for longer exposure times apart from the small overhead between successive images, described in Section 8.2