We summarize the results of the enabling and calibration activities involved in the first COS FUV move to a new lifetime position, which occurred on July 23, 2012.
R. Osten 08 Oct 2013
This document describes the results derived from the COS calibration Program 12676, "COS/FUV Characterization of Detector Effects".
D. Massa 01 Oct 2013
In this report, we describe the analysis we performed to verify the dispersion solutions of the M gratings using data collected in 12805 and of the L grating using data obtained in enabling program 12796 (FENA3:“Second COS FUV Lifetime Position: Focus Sweep Enabling Program”; Oliveira et al. 2013).
P. Sonnentrucker 10 Jun 2013
In this ISR we characterize the spatial and spectral resolution at the second COS FUV lifetime position.
J. Roman-Duval 10 Jun 2013
The document describes the analysis of data taken as part of the second lifetime calibration program 12807. This program was designed to characterize the BOA spectroscopic performance at the second lifetime position.
J. H. Debes 22 May 2013
This document summarizes the status of the calibration programs for COS during Cycle 18.
G. Kriss 26 Apr 2013
This document describes the analysis of the data obtained for proposal 12795 which determined the exact updates for the Science Instrument Aperture File (SIAF) and for the COS aperture block positioning needed to implement routine science operations at COS FUV Lifetime Position 2.
C. Proffitt 08 Apr 2013
This ISR reviews the causes of the shutdown program 12096 and how program 12677 was crafted to avoid a similar shutdown. The document presents results from program 12677 and their impact on future lifetime position is discussed.
C. Oliveira 07 Feb 2013
This document describes the results from the second lifetime position focus sweep with COS/FUV G130M, G160M, and G140L gratings as part of the programs executed to enable COS/FUV science at the second lifetime position.
C. Oliveira 06 Feb 2013
This document summarizes the status of the calibration programs for COS during Cycle 17.
R.Osten 29 Nov 2012
Describes the procedures for determining the COS FUV fine-focus for the G140L grating.
P. Ghavamian 03 Oct 2012
This ISR describes the behavior and characteristics of the COS FUV detector dark rate during SMOV and Cycle 17.
D. Sahnow 07 Oct 2011
SPIE paper describing the characteristics and mitigation strategies of gain sag in the FUV detector.
David J. Sahnow, Cristina Oliveira, Alessandra Aloisi, Philip E.
Hodge, Derck Massa, Rachel Osten, Charles Proffitt, Azalee
Bostroem, Jason B. McPhate, Stéphane Béland, Steven N. Osterman
and Steven V. Penton, "Gain sag in the FUV detector of the Cosmic
Origins Spectrograph", Proc. SPIE 8145, 81450Q (2011);
Copyright 2011 Society of Photo-Optical Instrumentation Engineers.
One print or electronic copy may be made for personal use only.
Systematic reproduction and distribution, duplication of any
material in this paper for a fee or for commercial purposes, or
modification of the content of the paper are prohibited.
D. Sahnow 26 Sep 2011
Describes the characteristics of COS TAGFLASH exposures.
T. Keyes 07 Sep 2011
Results of the FUV Gridwire Flat Field Development and Testing.
J. Ely 05 Aug 2011
Updated Results from the COS Spectroscopic Sensitivity Monitoring Program.
R. Osten 28 Jun 2011
This ISR describes the behavior and characteristics of the COS NUV detector dark rate during SMOV and Cycle 17.
D. Sahnow 12 Jul 2011
This ISR provides a description and analysis of the structural and thermal stability of the COS instrument.
T. Ed Smith 12 Jul 2011
We have developed an improved line spread function (LSF) for the medium resolution far- ultraviolet gratings (G130M and G160M) of the Cosmic Origins Spectrograph (COS) by taking into account scattering in the far wings due to microroughness on the HST primary mirror.
G. Kriss 18 Mar 2011
Presenting data analysis to describe NUV Imaging Performance for COS after SM4. Revised 10/6/2010
P. Goudfrooij 11 Mar 2010
Describes the external spectroscopic performance of the COS NUV detector
S. Béland 30 Sep 2010
A description of the External Spectroscopic Performance of the FUV Channel on COS
P. Ghavamian 30 Sep 2010
We report initial results from the Cosmic Origins Spectrograph spectroscopic sensitivity monitoring calibration programs.
R. Osten 02 Aug 2010
This ISR provides an overview of COS TA and updates, expands, and supersedes information in the COS IHB(v2)
T. Keyes 15 Jun 2010
The procedures for determining the COS FUV fine-focus are described
D. Lennon 04 Mar 2010
Update to the NUV wavelength calibration from SMOV
C. Oliveira 19 Feb 2010
Update to the FUV wavelength calibration from SMOV data
C. Oliveira 19 Feb 2010
SMOV NUV optical alignment process description
G. Hartig 12 Feb 2010
Describes the processes used to determine the absolute flux calibration of COS NUV point source spectra
D. Massa 28 Jan 2010
Describes the processes used to determine the absolute flux calibration of COS FUV point source spectra
D. Massa 28 Jan 2010
Describes the process of determining and results of the COS NUV high S/N flat field reference files from data taken during SMOV4
T. Ake 04 Feb 2010
Description of the COS Line Spread Function (LSF) as determined by post-launch testing.
P. Ghavamian 01 Oct 2009
Summary of recommended practice and suggestions for observers preparing their Phase II COS program submissions for Cycle 17.
D. Soderblom 06 Jun 2008
COS Bright Object
C. Leitherer 07 Mar 2006
COS Design Reference Mission and Ground System Volume Requirements
Charles (Tony) Keyes 07 Mar 2006