lists the estimated overheads associated with guide star acquisitions, the initial configuring (standby to operate) of the FGS for science observing, the time for slewing the IFOV from target to target in the observing plan, the acquisition of target stars, and finally configuring the FGS at the end of the observing sequence (operate to standby).
The acquisition time of an astrometric target depends upon the target’s magnitude. Fainter stars require longer intergration times (FESTIME) of the fine error signal during the walkdown to finelock. For example, the FESTIME used to acquire a V=10 star is 25 msec, while for a V=16 star FESTIME=3.2 seconds. Columns 2 and 3 in Table 6.6
list an average overhead for each exposure and the recommended total exposure time to be entered on the phase2 proposal (see Table 4.5
) as a function of target magnitude. Total time
required for a given exposure is estimated by combining columns 2 and 3. The additional overheads indicated in Table 6.5
must be included when determining the total time available for collecting science data. Fortunately, APT calculates and includes all of these overheads when processing a phase2 proposal.
The overheads for the activities which occur during Trans mode exposures are listed in Table 6.7
. The guide star acquisitions, instrumental configurations at the beginning and end of the observing sequence, and astrometry target acquisition activities for Pos mode and Trans mode are identical through the CoarseTrack phase. The FineLock phase in Transfer mode does not depend upon the star’s magnitude. (The star is not really acquired in FineLock. Setting the FineLock flag allows for the transition to fringe scanning under the control of the 486 SSM computer.)