ETC News and Known Issues
Be sure to read the Release Notes for new features and resolved issues in the ETC 25.2 release.
Some web browsers have a bug in which using the Back button causes ETC form values to reset. More information can be found in the User's Guide.
ETC Calculation Dates
ETC Calculations are currently performed with MJD = 58300.1 (July 1, 2018)
Contact the Helpdesk for access to Cycle 18 or prior
The Cycle 17 and Cycle 18 Java ETC is no longer available. Instead old calculation results from Cycles 17 and 18 have been archived. Contact the STScI Helpdesk at firstname.lastname@example.org in order to retrieve any old calculations necessary.
Using the ETC with archival data
There are a number of issues involved in using the ETC to support archival research. In particular,
- The ETC is optimized for projected instrument sensitivity and status during the upcoming observing cycle, and will thus supply incorrect answers for past instrument performance.
- The default SYNPHOT and pysynphot data files — graph (TMG), component (TMC), and thermal (TMT – for NICMOS and WFC3) tables are also optimized for projected use, and will thus also supply incorrect answers for past instrument performance.
- ETC calculations are not required to have the same standard of reliability and absolute accuracy with respect to sensitivity as instrument calibration files and pipelines. In particular, for instruments with health and safety concerns the ETC provides a conservative estimate.
The following issues remain unresolved in 25.2:
- An invalid value in an input field, even a field for an option that is not selected,
will cause the ETC to return with an Input Error.
An invalid value in an input page field (say, a negative extraction region radius), even when that field is not selected, still triggers the checks for validity in the web server. This causes error pages to be displayed even though the invalid field is not selected.
- ETC User's Guide Display Issues
On the ETC User Spectra page, the units will appear garbled if the character encoding is not correctly set to Unicode. The Firefox browser defaults to ISO-8859-1 (Western) encoding on this page. Manually setting the encoding to Unicode via View —> Character Encoding will fix the issue.
- ETC Results Plot Failure when flux or throughput is uniformly zero
When an ETC result is returned with an observed flux of zero or throughput of zero, the related ETC plot will return an error rather than being displayed.
- ACS results with CLEAR do not include known effects of PSF degradation
Filterless ACS observations (obtained by setting both filters to "CLEAR") are available but not supported by ACS calibration programs. Such observations are out of focus, resulting in a broader PSF (less encircled energy) than otherwise expected, as documented in ACS ISR 2003-03. This PSF degradation is not included in the ETC calculations, and the ETC will thus tend to overestimate SNR in those cases.
- It is possible to enter a “Post Flash” value in SBC observations.
The ETC will allow a value to be entered for post-flash in SBC imaging observations. The ETC will then ignore this value, as post flash does not apply to the SBC.
NICMOS ETC No Longer Available
The NICMOS ETC is no longer available for new calculations. Given the fact that NICMOS is not being offered for use in the current proposal cycles, and due to the substantial maintenance and development costs in a constrained budgetary environment, the decision was taken to discontinue support for the NICMOS ETC.
Users can still retrieve NICMOS ETC results from Cycles 17 and 18 by contacting the STScI Help Desk.
In order to (for example) calculate the effective wavelengths of NICMOS filters, pysynphot may be used, provided that the Throughput Tables have been downloaded. For example, to calculate the effective wavelength of the NICMOS f222m filter,
>>> import pysynphot as S >>> sp=S.FileSpectrum(’$PYSYN_CDBS/calspec/gd71_stisnic_003.fits’) >>> bp=S.ObsBandpass(‘nicmos,3,f222m’) >>> obs=S.Observation(sp,bp) >>> obs.efflam() 22150.506378238068For assistance with pysynphot, please contact the helpdesk.
Option for in-band normalization
The option for in-band normalization currently is not supported for certain STIS modes, in particular for the 50CCD filter in imaging modes and for slits that have numbers followed by one or more letters (e.g. 0.2x0.6). Performing such calculations will cause the ETC to produce an error.
Error in brightest pixel count rate for observations with 52x0.05 slit
In cases where the brightest pixel geometry is less than one pixel in size due to aperture effects, the ETC code will ignore the effects of the aperture and calculate the brightest pixel as a full pixel. This will result in the brightest pixel count rate being overestimated.
Buffer Time in Previous ETC Results
Older releases of the ETC did not calculate the STIS MAMA buffer times, so recalling previous results will produce the text "ERROR, number expected." Users wishing to determine the buffer time should re-run their calculations in the current release of the ETC.
Wrong CCD chip selection for WFC3/UVIS quad filters is permitted
Each WFC3/UVIS quad filter (named F*Q) covers somewhat less than a quadrant of the UVIS FOV on a particular chip, 1 or 2. Even though the ETC will let the user select either chip, when only one is physically possible, the ETC will now compute the correct results, using the chip associated with the quad filter regardless of user input.
Option for in-band normalization
The option for in-band normalization currently is not supported for certain WFC3 modes, in particular for the 50CCD filter and any mode which begins with a number but contains letters.
- WFC3IR Imaging has a limited range of values for ‘Percentage of all light’
In the WFC3 IR Imaging ETC with default settings, when ‘User a circular region containing X percent of total light’ is chosen, there is not sufficient data to handle values less than 50% or greater than 99.9%. Selecting such values will result in a ‘Value out of range’ error.
- Default Aperture for WFC3 (UVIS and IR) is 0.2 arcseconds
Please note that the default aperture radius for WFC3 is 0.2 arcsec. This aperture includes ~84 to 86% of encircled energy for wavelengths in common to WFC3/UVIS and ACS/WFC, and ~76 to 80% of encircled energy for wavelengths in common to WFC3/IR and ACS/WFC. (See Tables 6.8 and 7.6 in the WFC3 Instrument Handbook for WFC3/UVIS and WFC3/IR, respectively.) For direct comparison to the ACS default aperture, use a circular region with 80% of total light. Note that this option will not work in all cases.
- Chrome browser behaves badly with back button for WFC3 stare and scan spectroscopic modes
When disperser G141 has been selected, the simulation is run, and the option to return to the previous page is used in the Google Chrome browser, the input parameters are preserved but there is an inappropriate error message on the input line for the vacuum wavelength (Error: wavelength is not in the passband) and the message below that input line is “Filter G102 pass band 0.8 – 1.1645” instead of “Filter G141 pass band 1.0665 – 1.7205”. The next simulation will fail if the wavelength error message is present. It can be removed and the appropriate pass band statement restored by selecting G102 then G141 as the disperser.