Space Telescope Science Institute STIS Instrument Handbook help@stsci.edu

Space Telescope Imaging Spectrograph Instrument Handbook for Cycle 21 > Chapter 6: Exposure Time Calculations > 6.3 Throughput and Sensitivity

 6.3 Throughput and Sensitivity
So far, we have given the formulae for count rates in terms of the sensitivity (S). If you would like to use the throughput (T) rather than the sensitivity, you can use the following conversion equations.
For imaging modes, the transformation between T and S is given by:
where:
 • Sλ is the sensitivity at wavelength λ;
 • Tλ is the system throughput at λ;
 • A is the area of an unobstructed 2.4 meter telescope (i.e., 45,239 cm2);
 • h is Planck’s constant;
 • c is the velocity of light;
 • The effective area Aeff is given by Aeff = TλA.
For spectroscopic modes, the transformation equation can be written as:
where:
 • d is the dispersion in Å/pix.
In the first two examples given at the end of this chapter, we specifically show how the calculations can be done both in terms of the sensitivity and the throughput.