The UVIS channel employs a mosaic of two 4096 ×
(formerly Marconi Applied Technologies Ltd.) CCDs, with ~0.040"/pixels, covering a nominal 162 ×
162" field of view. These CCDs are thinned and back-illuminated devices cooled by thermo-electric cooler (TEC) stacks and housed in sealed, evacuated dewars with fused silica windows, nearly identical to the ones used for ACS. The spectral response of the UVIS CCDs is optimized for imaging from the near-UV at 200 nm to visible wavelengths at 1000 nm. The two CCDs are butted together but have a 35-pixel gap between the two chips (~1.4" on the sky). Minimum UVIS exposure time is 0.5 sec. The dynamic range for a single exposure is ultimately limited by the depth of the CCD full well (~70,000 electrons), which determines the total amount of charge that can accumulate in any one pixel during an exposure without saturation.
WFC3 recycles hardware used in WF/PC-1 to house the complement of filters for the UVIS channel. The Selectable Optical Filter Assembly (SOFA) contains a stack of 12 wheels housing 48 physical elements covering the UV/Visible range: 42 full-frame filters, 5 quad filters (2 ×
2 mosaics providing 4 different bandpasses), and 1 grism, giving a total of 63 spectral elements. Each wheel has an open slot such that when an observation is taking place, the appropriate wheel is rotated to place the desired filter in the beam, while the other wheels place the open slot in the light path.
shows a schematic of the UVIS channel aperture projected onto the sky with respect to the U2/U3 reference frame. (For definitions of the coordinate systems in the figure, please refer to Section 7.4.3 of the
WFC3 Instrument Handbook
.) The WFC3 optics cause the nominally square field of view of the UVIS detector to be projected onto the sky as a skewed rhombus, 162" on a side, with an angle of 86.1 degrees between the sides. This distortion affects both the photometric accuracy and astrometric precision of the UVIS images. For a thorough discussion of WFC3’s geometric distortion, we refer the reader to
illustrates the fiducial points of the full-detector apertures (UVIS, UVIS1, UVIS2, and UVIS-CENTER), and the outlines of the 2Kx2K, 1Kx1K, and 512x512 subarray apertures. Also indicated are the positions of the four readout amplifiers (A, B, C, and D). The regions imaged by the UVIS detector (represented by blue fill) and by the IR detector (represented by red fill) are also indicated. The POSition TARGet coordinate system for the UVIS-CENTER aperture, with its origin at that aperture's fiducial point, is illustrated. Although the POSition TARGet coordinate systems for the other apertures are not illustrated, they are oriented the same, but have origins at each aperture's fiducial point. (U2 = –V2 and U3 = –V3).