WFC3 Data Handbook v.4 |

help@stsci.edu |

WFC3 Data Handbook > Chapter 7: WFC3/IR Sources of Error > 7.2 Gain

7.2 The WFC3 IR detector has a nominal gain setting of 2.5 e-/DN in order to provide complete sampling of the full-well of the IR detector (~80,000 electrons, which is ~32,000 DN at the nominal gain). Due to slight hardware differences, effective gain values in each quadrant of the detector deviate slightly from the nominal gain. Twice yearly, the effective gain is measured via the mean-variance technique on pairs of internal flat fields: the mean signal level is plotted against the variance in signal, and the gain is the inverse of the slope. See WFC3 ISR 2015-14 for more information on this method and the status of the IR gain from 2010-2015. Table 7.1 summarizes the current values of the effective gain measured in each quadrant of the IR detector. The WFC3 group will continue to monitor the effective gain of the detector twice yearly to evaluate its stability over time.

WFC3 Data Handbook > Chapter 7: WFC3/IR Sources of Error > 7.2 Gain