Following its alignment to the OTA (WFC3 ISR 2009-46
), a series of observations through a variety of filters were obtained to demonstrate the WFC3 optical performance. The WFC3 IR channel meets or exceeds all image quality expectations. The following subsections summarize the measured flight optical performance for the IR channel, as described by its point-spread function (PSF), i.e., the spatial distribution of the flux in an image of a point source. The results shown are produced using an optical model which has been adjusted and correlated to the PSFs measured on-orbit and represent mean values averaged over the field. (See WFC3 ISR 2009-37
.) This PSF model includes the pupil geometry, including the cold stop, residual aberration, the mid-frequency wavefront error of the OTA, the effect of the detector inter-pixel capacitive cross-talk, and first-order geometric distortion.
The IR channel PSFs are well approximated by Gaussian profiles (before pixelation). As was discussed in more detail for the UVIS channel in Section 6.6.1
, the PSFs can usefully be characterized by their FWHM or their sharpness (effectively the reciprocal of the number of pixels occupied by a point source, as defined in Section 6.6.1
). Table 7.5
lists the FWHM of the model PSF core (i.e., before pixelation) in units of pixel and arcsec and the sharpness parameter for 10 wavelengths. The sharpness range shown for each wavelength indicates the values for the PSF centered on the pixel corners and center. The monotonic increase in FWHM and decrease in sharpness with wavelength is due to diffraction.
plots the azimuthally-averaged model OTA+WFC3 PSF at three different IR wavelengths., indicating the fractional PSF flux per pixel at radii from 1 pixel to 5 arcsec.
As described in more detail in Section 6.6.2
for the UVIS channel, encircled energy, the fraction of light contained in a circular aperture, and ensquared energy, the fraction of energy falling within a certain number of pixels, are two other useful ways of characterizing well-behaved profiles.
Encircled-energy profiles for the IR channel at three representative wavelengths are plotted in Figure 7.5
. The corresponding numerical values are presented in Table 7.6
. The ensquared energy, which provides the flux per pixel, is presented in Table 7.7
During SMOV, high-dynamic-range isolated star images were obtained through several filters (WFC3 ISR 2009-37).
These are shown with a logarithmic stretch in Figure 7.6
. The images appear slightly elongated vertically, due to the 24 degree tilt of the detector to the chief ray and the fact that a distortion correction has not been applied. Although the target was chosen to be isolated, a number of field galaxies appear in both the F098M and the F160W filter images; these are also seen in long wavelength UVIS channel images of the same target (Figure 6.13
). Some detector artifacts, including cold and warm pixels and imperfectly removed cosmic ray hits are evident.
observations are affected to some extent by short-term focus variations (“breathing”), which arise from the changing thermal environment of HST
(See Section 6.6.3
). For the IR channel, breathing is expected to produce variations of ~0.3% and less than 0.1% in the FWHM of the PSF of WFC3/IR at 700 nm and at wavelengths longer than 1100 nm, respectively, on typical time scales of one orbit.
For the WFC3/IR flight detector, no measurable intra-pixel sensitivity variation was found during ground testing (WFC3 ISR 2008-29
) or in-flight testing (WFC3 ISR 2011-19
). The smaller pixel size (18 µm vs. 40 µm) and the much higher WFC3/IR detector temperature (compared to NICMOS) are probably responsible for this improvement.
The small pixel size, relative to that in NICMOS, increases the relevance of capacitive coupling between nearby pixels (see Brown et. al., 2006, PASP
, 1443; Moore et. al., 2006, Opt. Eng., 076402). It affects the gain measurements and the PSF. The easiest method of estimating the inter-pixel capacitance is to measure the ratio of DNs in pixels adjacent to a “hot” (high-dark-current) pixel to the DNs in the hot pixel. In the WFC3 IR channel, on the order of 5% of electrons may leak to the adjacent pixels. WFC3 ISR 2008-26
describes a method for correcting inter-pixel capacitance using Fourier deconvolution, and demonstrates its effectiveness on WFC3/IR ground-testing data. WFC3 ISR 2011-10
provides an improved deconvolution kernel with distinct values for each of the 8 pixels surrounding the central pixel.