|Space Telescope Science Institute|
|WFC3 Instrument Handbook|
9.4.1 Point SourceFor a point source, the count rate, C (e− s−1), can be expressed as the following integral over the bandpass of the filter:
• A is the area of an unobstructed 2.4-m telescope (i.e., 45,239 cm2)
• Fλ is the flux from the astronomical source in erg cm−2 s−1 Å−1.
• The factor λ/hc (where h is Planck’s constant and c is the speed of light) converts ergs to photons.
• QλTλ is the system fractional throughput, i.e., the probability of detecting an electron per incident photon, including losses due to obstructions of the full 2.4-m OTA aperture. It is specified this way to separate out the instrument sensitivity Qλ and the filter transmission Tλ.
• εf is the fraction of the point-source energy encircled within Npix pixels.
• Sλ is the total imaging point-source sensitivity in units of e− s−1 Å−1
per incident erg cm−2 s−1 Å−1.The peak e− s−1 pixel−1 from the point source, Cpeak, is given by the following integral over the bandpass:
• Fλ, and Sλ are as above.
• εf(1) is the fraction of energy contained within the peak pixel.If the flux from the source can be approximated by a flat continuum (Fλ = constant) and εf is roughly constant over the bandpass, then:We can now define an equivalent bandpass of the filter, Bλ, such that:
• Speak is the peak sensitivity.
• Bλ is the effective bandpass of the filter.where V is the visual magnitude of the source, the quantity under the integral is the mean sensitivity of the detector+filter combination (also tabulated in Tables 9.1 and 9.2), and ΑΒν is the filter-dependent correction for the deviation of the source spectrum from a constant Fν spectrum. This latter quantity is tabulated for some representative astronomical spectra in Appendix A:WFC3 Filter Throughputs.9.4.2 Diffuse SourcesFor a diffuse source, the count rate, C (e− s−1 pixel−1), which is now per pixel, due to the astronomical source can be expressed as:
• Iλ is the surface brightness of the astronomical source, in
erg cm−2 s−1 Å−1 arcsec−2.
• Sλ is as above.
• mx and my are the plate scales in arcsec pixel−1 along orthogonal axes.9.4.3 Emission-Line SourcesFor a source where the flux is dominated by a single emission line, the count rate can be calculated from the equationwhere C is the observed count rate in e− s−1, (QT)λ is the system throughput at the wavelength of the emission line, F(λ) is the emission-line flux in units of erg cm−2 s−1, and λ is the wavelength of the emission line in angstroms. (QT)λ can be determined by inspection of the plots in Appendix A:WFC3 Filter Throughputs. See Section 9.9.4 for an example of count-rate estimation for an emission-line source.